S-100L double-layer glass reactor product introduction
The reaction solvent can be stirred in the inner layer of the S-100L double-layer glass reactor, and the interlayer can be circulated through the cold heat source (refrigerant, water, heat transfer oil), so that the inner layer can be heated or cooled, and the reaction solvent can be controlled. Distillation and reflux are ideal test and production equipment for modern synthetic chemicals, biopharmaceuticals and new materials.
The S-100 double-layer glass reactor can realize the multi-purpose of the experiment and meet the needs of various aspects of different experiments. The equipment can be carried out: vacuum stirring reaction, high and low temperature reaction, constant speed running mixing reaction, distillation, reflux, Concentration and other test operations, our company can be assembled into a glass reactor production line according to the specific requirements of customers, according to the different requirements of customers for non-standard processing custom.
S-100L double-layer glass reactor product details
Technical characteristics of S-100 liter double-layer glass reactor
· The whole stainless steel column mobile frame structure, five reactor lids, complete glass with reflow, liquid addition and temperature measurement.
· G3.3 borosilicate glass is used, which has good chemical and physical properties.
· Alloy steel mechanical seal, Teflon joint, high precision seal under working condition.
· Pt100 sensor probe has high temperature measurement accuracy and small error, which effectively improves work efficiency.
· Japanese technology AC gear reduction motor, strong torque, no noise.
· Double PTFE stirring paddle for mixing and mixing of low to high viscosity liquids.
· PTFE discharge valve, movable interface, complete and quick discharge.
S-100 liter double-layer glass reactor step
1. After opening the package, check whether the main accessories of the machine are complete according to the packing list.
2. Assemble the frame in accordance with the drawings as shown in the figure.
3. Install the electrical box on the top of the right rear pole and tighten the screws and insert the seven-pin plug. Install the vacuum gauge on the left rear end and tighten the screws.
4. According to the height of use, fix the round bracket of the kettle to the vertical pole slider, and place the kettle on the bracket. The semi-circular hoop is used to insert the pole of the kettle into the pole of the kettle, and then tighten the fixing screws after closing. Pay attention to the vertical direction of the reactor body during installation.
5. The stir bar is fixed on the toothed ring chuck of the motor main shaft. The stir bar passes through the middle rotating bearing of the cover, tightens the special connector, and then adjusts the position of the motor, paying attention to the vertical concentricity.
6. The slider on the pole is a stainless steel multi-purpose clamp that adjusts the direction and height in different directions.
7. On the left side of the bottle cap, the 40# standard port plug-in type reflux condenser, the right side 40# is the feeding port connected to the constant pressure funnel, the middle front 24# standard port is inserted into the thermometer casing mouth, and the rear of the hat is the 34# standard port. It is a multi-functional spare port, and the bottom is provided with a discharge valve, and the bottom of the kettle body is respectively a circulating fluid inlet and outlet. The lower port is connected to the circulating fluid inlet, and the upper port is connected to the circulating fluid return port.
8. When installing the glass instrument, it must be cleaned, and the interface should be coated with Vaseline to prevent the glass from locking up, and then apply vacuum grease to prevent air leakage.
9. Press the universal wheel wheel fixing device to stir. If the stirring is stable, the debugging is good.
S-100 liter double-layer glass reactor operation procedure
1. Check that the power supply voltage is consistent with the specifications provided on the machine nameplate.
2. Plug in the power plug, turn on the power switch on the inverter, and use the speed knob to select the appropriate speed.
3. The flow of material and the power of the motor speed may resonate at a certain point. Please change the speed of the motor to avoid resonance.
4. After the stir bar is installed, it must be rotated by hand. Pay attention to whether the concentricity is good. If the heart is not good, the heavy clamp should be loosened. After the clamp is turned on, turn on the power and adjust gradually from slow to fast.
5, using the stir bar regulator, you can make the stirring leaves up and down 5cm as needed.
6. If the seal performance of the kettle is found to be degraded, please check the rotary bearing in the stirrer sleeve.
7. The glass equipment of this reaction device is all made of GG.17 low expansion coefficient high temperature resistant borosilicate glass. The user should self-fit glass bottles such as reaction bottles. Please confirm the quasi-glass material to ensure safe use.
8. The external layer of the interlayer on the glass kettle can be heated by the hot oil. The coolant can be used for low temperature reaction. If it is heated by steam, it should not exceed 0.5K-1K. (Try not to use pressure steam to heat).
9. When the double-layer glass reactor is tested at low temperature, the discharge valve at the bottom will be frosted. When using the valve, it must be partially thawed to prevent the glass from being broken.
10. If there are particulate objects in the solution during use, it is possible to store the remaining materials on the PTFE piston of the valve. When used again, the airtightness will be affected. Be sure to clean and re-use each time after discharging. .
S-100 liter double-layer glass reactor maintenance
A. Check the instrument carefully before use. Whether the glass bottle is damaged or not, and whether the interfaces are consistent, pay attention to light handling.
B. Note that this equipment is a fragile product. Please be careful when using it. The damage of fragile goods is not covered by our company's warranty. If the customer needs it, we will charge the corresponding cost.
C. The PTFE switch can not be tightened too much, which is easy to damage the glass. 4. Turn on the power switch first, then let the machine run from slow to fast. When the machine is stopped, stop the machine and turn off the switch.
D. Each interface can't be tightened too tightly. It should be loosened regularly to avoid the long-term tight lock and cause the connector to die.
E. Every time you use it, you must use a soft cloth to clean all kinds of oil traces left on the surface of the machine, stains, solvent remaining, and keep it clean.
F. Clean the seal ring regularly by removing the seal ring and checking if there is dirt on the shaft. Wipe it with a soft cloth, then apply a little vacuum grease and reinstall it to keep the shaft and seal ring smooth.
G. Loosen the PTFE switches after the machine stops. The long-term stationary state will make the PTFE piston deform.
H, the electrical part must not enter the water, it is strictly prohibited to get wet.
I. The original authentic accessories must be purchased. Any use of other accessories can cause fatal damage to the machine.
J. When doing any repairs or inspections on the machine, be sure to cut off the power supply and water source.
K. The instrument is returned to the manufacturer for maintenance once after three years of use (recommended).
L. Wipe each interface with a soft cloth (can be replaced with a napkin) and apply a small amount of vacuum grease. (The vacuum grease must be covered after use to prevent the entry of sand.)
GET A QUOTE